PROCEDURE TO CONNECT THE LIMIT SWITCHES
Fig 1. This shows the connection of the Trip pin on the motor control box. This pin provides hardware limit switches to restrict the movement of the diffractometer arms in a certain range.
Fig 4. This picture
shows another limit switch on the diffractometer. This switch serves to
restrict the Chi circle movement to a particular range.